KEYENCE LK-GD500 USB DRIVER DOWNLOAD
Incoming command Reception Response Response command t The t-time is identified as follows. Page 50 2 Operations and Functions during Measurement Note Measurement stops while the setting is in progress. Page 61 3 Function Settings Setting the Head This section describes the settings related to the sensing for stable detection. Or point us to the URL where the manual is located. The ABLE function automatically adjusts the light intensity and the sensitivity to their optimum levels according to the conditions color, luster, and material of the target surface.
|Date Added:||4 December 2014|
|File Size:||66.94 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Measurement value Input waveform 3 Time Measurement value Measurement value Waveform by filtering with high-pass. Page 48 2 Operations and Functions during Measurement Switching the Measurement Value Displays This section describes how to switch the measurement value display contents. Route the display panel cable along the guide, and connect it to the display panel connector on the rear of the controller. Page 61 3 Function Settings Setting the Head This section describes the settings related to the sensing for stable detection.
Set the OUT No. Set the scaling of the analog output. Page 98 3 Function Settings Setting the Mutual Interference Prevention Function When two heads are connected and a little distance is left usn them, a measurement error may occur by receiving the light emitted from the other head. Be sure that the person who will finally operate this product receives this manual.
Change the setting as required. Releases the holding when the measurement value is reset. K eep this manual in a saf e place f or future. Press the key, and select “A-2” with the key.
Refer to “Outline of the command format” page for the error response command. List of Functions and Function Nos. Binary output cannot be Is the unit properly wired?
For changing the OUT No. Refer to page Press the key, and select function “k” with the key.
Press the key, and select “d-1” with the key. The current measurement value is loaded if the key is pressed. Note These commands are accepted only when the controller is in the “Normal mode”. Sets the number of times of averaging.
The l-gd500 return to the default setting when shipped from the factory. Note Insulated mounting is recommended when using the head under the positive grounding environment.
The data to be used for the statistical computation is the one held in each measurement mode. Receives the measurement control command. Set the alarm process.
Refer to “Outline of the command format” page for the response command when an abnormality has occurred. Function Function Function Function No. Hysteresis is not set by default when shipped from the factory.
The alarm process specifies how to process the data in these situations. The measurement value display becomes zero.
Specifications : LK-G series | KEYENCE Philippines
Display Panel Two protrusions Display panel fixing case optional 2 Fix the display panel with the panel attachment ring, and connect the cm display cable optional. Influence of dust or dirt The measurement may fluctuate due to dirt, dust or fluid such as water or oil in the following cases: Use this function when the reflective ratio of the target changes drastically in a short cycle.
This function can be used for reference zero-point positioning when a workpiece is changed.