MEDTRONIC IPRO2 DOCK DRIVER
This is normal because the iPro2 has not been activated. For example, you may need to change a user password, or grant or remove administrative privileges. Use of the continuous glucose monitoring system to guide therapy in patients with insulin- treated diabetes. If connected to an electrical outlet, it is ready to charge an iPro2. Docking Station Dock – Device that performs two functions: Wait for the Dock to show a solid green charging light.
|Date Added:||18 July 2016|
|File Size:||35.76 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Medtronic Ipro2 Dock driver direct download was reported as adequate by a large percentage of our reporters, so it should be good to download and install.
The red warning light may turn on if you do not immediately complete the next steps. The highest MAD for adult subjects Page 5 warnings 4 viewing Logbook entries 48 patient reports 50 wall-powered adapter about 2 warnings 4 warranty 91 water-tightness Dock 62 where to insert sensor 29 white light 74 wireless devices, interference 8 iPro2 CGM User Guide Index If the iPro2 is disconnected from the sensor and then reconnected during the study, it will continue recording.
Drivers for Medtronic iPro2 Dock
Deleting User Accounts Select the user account and click the Open user button. Page 3 Medtronic B.
Identify priority issues Top three issues prioritized Visuals of meal times and daily glucose values. Don’t have an account? If you want to apply the new Report Settings for the patient to one or more studies, find the study on the patient record screen medtrinic select Regenerate reports from the Other Options drop-down list.
Fluids can cause the connector to corrode and affect the iPro2’s performance.
Page 60 BG meter readings with which to calibrate the sensor values. Got it, continue to print. Restoring Medttonic Default Report Settings If you want to apply the new Report Settings for the patient to one or more studies, find the study on the patient record screen and select Regenerate reports from the Other Options drop-down list.
In the event a iPro2 or Dock is repaired or replaced, the warranty period will not be extended past its original expiration date. If the green charging light on the Dock is on and no longer flashing, the dick is charged and ready for the next patient.
Medtronic iPro2 Dock Drivers
This can help you find items in the Logbook. Medtronic Ipro2 Dock driver installation manager was reported as very satisfying by a large percentage of our reporters, so it is recommended to download and install.
If you connect the Dock before the iPro2 Dock driver is installed, it may prevent the iPro2 Dock driver from installing properly.
Moisture inside the connector could cause the iPro2 to not work properly, and could cause corrosion and damage over time. We hope you will find iPro2 to be the simplest and most convenient CGM product that you have ever used. If the patient does the first BG meter reading too soon, sensor data will not be available for calibration. Are not at their HbA1c target Hypoglycaemia Believe they are experiencing the effect of hypo unawareness Are changing their diabetes therapy Are undergoing lifestyle challenges.
Each study has its own set of reports. Typically, candidates for Professional CGM are patients with Type 1 or Type 2 diabetes who experience any of the following:.
Medtronic iPro2 User Manual
Viewing And Printing Patient Reports Keep in mind that you need BG meter readings within 12 hours of each other in order to avoid data gaps. Comments to this Manuals Your Name. Page Docking Station Dock – Device that performs two functions: The customer or users of the All components of the iPro2 CGM system are suitable for use mdtronic a clinical environment.
I have read and understand the terms of this marketing authorization. Page 94 The iPro2 is an internally powered device.
Uploading Data To Carelink Ipro Replace the cleaning plug after 30 uses to maintain a water-tight seal. Where To Insert The Sensor Make sure that your patient tests blood glucose at least one hour after the iPro2 is connected to the sensor.
Msdtronic consent to receiving text messages at the phone number s submitted above using automated technology from or on behalf of Medtronic MiniMed, Inc.